Masimo tests and validates a variety of smartphones to ensure compatibility. If you do not see your smartphone currently listed below, check back soon, as we frequently add support for new devices. Please contact Masimo Customer Support with any questions.
Tablets are not compatible with Masimo SafetyNet Alert at this time.
The Masimo SafetyNet Alert kit contains two (2) Masimo Sensors, one (1) Masimo Chip, 12 replacement tapes, one (1) Bedside Station, a User’s Guide, and Quick Start Instructions. The Masimo SafetyNet App is required and available on the App Store for iOS devices and Google Play for Android devices.
Search for “Masimo SafetyNet Alert” on the App Store or Google Play.
Yes, a smartphone equipped with Bluetooth® is necessary to use Masimo SafetyNet Alert.
Masimo SafetyNet Alert is compatible with most Android and iOS devices.
No, Masimo SafetyNet Alert is not currently configured for use with tablets.
Yes, a Wi-Fi connection is necessary for the initial setup of the system and during monitoring. The bedside station must remain connected to Wi-Fi during monitoring for the system to work properly.
No, a user is not required to designate any Emergency Contacts to use Masimo SafetyNet Alert, but it is recommended.
No, an Emergency Contact does not need to download the app. Notifications are sent by SMS.
Yes, a text message invitation is sent to each Emergency Contact designated in the app. The Emergency Contact must accept the invitation by replying with the code provided in the invitation.
In the Emergency Contact section of the app, each Emergency Contact carries a status label indicating “Accepted,” “Pending,” or “Unsubscribed.”
A user can designate up to 10 Emergency Contacts.
Yes, each Emergency Contact receiving text notifications for a particular user will receive the same information.
Yes, a text notification is sent to the Emergency Contact indicating that they are no longer an Emergency Contact.
Get more information on Emergency Contacts.
Each sensor’s battery should last for up to 96 hours of continuous use. A flashing orange light indicates the sensor battery is low. A flashing red light indicates the sensor battery needs to be replaced.
When the sensor is not in use, remove the reusable chip and place the yellow adhesive label back on the chip holder portion of the sensor.
The bedside station must be within standard Bluetooth range of your sensor to receive data. This distance can vary depending on the use environment. Maintaining a direct line of sight between the sensor and bedside station is recommended.
If the sensor is out of range of the bedside station, the user will receive a banner message in the app. The sensor Light Indicator (on the chip) will change from blue to green when it is no longer Bluetooth-paired with the system.
Yes, the app can run in the background and continue to receive data collected from the sensor. Do not close/quit the app during a monitoring session.
You can wash your hands while wearing the sensor. However, avoid submerging it in water. Do not wear the sensor in the shower or bath. You should replace the adhesive tape when it is dirty or no longer sticking.
The Masimo sensor and adhesive tapes can be disposed of as part of standard household waste collection. The Masimo Chip can be reused and should not be discarded.
The Masimo Chip includes a light to indicate sensor and pairing status. Refer to the table below for a description of what the colors indicate. This information can also be found in the app under Main Menu > Devices > Sensor.
During active monitoring, the bedside station will alarm if it loses Wi-Fi connection.
Yes, if the phone has a data connection, the system will be able to send text notification to the cloud and to Emergency Contacts.
Masimo SafetyNet Alert does not support monitoring multiple individuals simultaneously. A complete system consisting of a chip, a sensor, a bedside station, and the app is required for each person. However, two or more systems will work simultaneously on the same wireless network.
No, the alert thresholds are fixed and cannot be modified by the user.
Location services is required during initial Wi-Fi setup and to reconnect a bedside station to Wi-Fi. It is recommended that location services remain active during monitoring.
The screen will display a “low pulse rate confidence” message and recommended actions.
Yes, monitoring history is available from the Main Menu.
Follow on-screen instructions for recommended actions. Clinical decisions are the responsibility of your doctor or other healthcare provider.
The system continues to monitor and send data to the cloud. If a text notification is triggered, the bedside station will send data to the cloud, which will trigger a text notification to your Emergency Contacts.
Yes, the system will continue to provide alarms at the phone and bedside station and continue to send text notifications to Emergency Contacts (if triggered).
You should use Masimo SafetyNet for as long as your care provider recommends.
Masimo SafetyNet Alert is intended for individuals weighing 40 kg or greater.
Low oxygen level alerts are triggered when oxygen levels fall below fixed thresholds. When these events occur, you are alerted with audible alarms that sound on the bedside station and your smartphone. If oxygen levels drop into Level 2 range, in addition to bedside and smartphone alarms, Emergency Contacts will receive a text message notification that your oxygen level is low.
|Oxygen level||Alarms||Text Notifications|
|Level 1 Less than 85% for 30 seconds or more||Audible alarms sound on smartphone and bedside station.|
|Level 2 Less than 80% for 30 seconds or more||Text message notification sent to Emergency Contacts.|
|Level 3 Less than 80% for 180 seconds or more; or less than 60% for 60 seconds or more|
Masimo SafetyNet Alert data is stored in the cloud and on the smartphone. Vital signs data reports are stored in the cloud. Any data entered manually is stored on your smartphone.
From the app, choose History from the Main Menu.
No. the Masimo SafetyNet Alert App does not currently support direct data sharing.
Third-party vendors do not have access to your personal information or data.
No, a user’s location is not stored and cannot be tracked by Masimo.
|Symptom||Potential Causes||Next Steps|
|Vitals data not displayed or measurement values don’t look correct||
|Unable to pair your sensor to your mobile device||
If you have tried all of these steps but still have issues, please contact Masimo Customer Support.
Masimo SafetyNet Alert has obtained CE Marking. Not available in all markets. Check with your sales rep or customer service for product availability.